No. Original manifolds do not align correctly on the new instrument.
Short fragments tend to create smaller clusters, allowing greater data density. The optimal fragment size for a single-read run is 150–300 bp. The optimal fragment size for a paired-end run is 250–500 bp.
The first time you process template DNA, try a concentration titration range to optimize the number of clusters formed. If the concentration is too low, fewer clusters are generated and result in a low sequencing yield. If the concentration is too high, clusters are too dense and can complicate data analysis. For more information, see the Sequencing Library qPCR Quantification Guide (requires a MyIllumina login) or the HiSeq Systems Denature and Dilute Libraries Guide.
cDNA libraries should be considered an equivalent to gDNA for clustering and sequencing steps, assuming an equivalent complexity of the sample.
For unbalanced libraries, Illumina recommends a high-concentration spike-in of PhiX control lane to create more diverse.
Unbalanced libraries contain genomes with AT or GC content of less than 40% or greater than 60%. PhiX Control v3 (catalog # FC-110-3001) is a diverse library with a mean insert size of ~375 bp, corresponding to a 500 bp library size if visualized on a Bioanalyzer.
For more information, see Using a PhiX Control for HiSeq Sequencing Runs.
Yes. The new cluster protocol and wider lanes on flow cell v3 make it imperative to re-titrate your samples to optimize cluster densities for higher throughput. qPCR is highly recommended.
For most libraries, Illumina recommends using a low-concentration
spike-in (1%) of PhiX Control v3 as an in-lane positive control for
alignment calculations and quantification efficiency.
After amplification, linearization, blocking, and primer hybridization, you can store the flow cell in storage buffer at 2° to 8°C for up to 10 days. HiSeq X flow cells can be stored up to 48 hours.
For rapid flow cells, perform the sequencing run on the same day as sample loading.
Not at this time. If you are sequencing Nextera libraries, you need sequencing primers provided in the TruSeq Dual Index Sequencing Primer Box for any run types. If you are sequencing dual-indexed TruSeq HT libraries on a single-read flow cell, you need sequencing primers provided in the TruSeq Dual Index Sequencing Primer Box, Single Read.
You need to increase your starting template concentration. However, the starting template volume (120 µl) is unchanged from previous kits.
No. The Illumina PhiX Control v3 is a separate product. The catalog number is: FC-110-3001.
Yes. The HiSeq flow cell requires the use of a cBot for clustering on the flow cell prior to sequencing.
On the Cluster Station, it takes about 5 hours. On the cBot, clustering duration depends on the flow cell that you are using. HiSeq rapid flow cells take about 1 hour to cluster, and other HiSeq flow cells take about 3 hours. TruSeq v3 and GAIIx v2 flow cells are clustered in about 5 hours.
No. Cluster generation is performed on the MiSeq System as part of the run. No other instruments are required for sequencing on the MiSeq.