The Custom Protocol Selector enables you to generate your own customized documentation, specifically tailored to your experiment on the NovaSeq 6000, MiniSeq, MiSeq, NextSeq, HiSeq 2000, HiSeq 2500, HiSeq 3000, HiSeq 4000, or HiSeq X systems. You select the products and workflow options, then the Custom Protocol Selector generates a PDF with only the instructions that you need.
This guide contains safety information and product compliance markings for the MiSeq System.
This guide provides requirements for preparing your lab for a MiSeq System: lab space requirements, electrical requirements, bench recommendations, UPS recommendations, and user-supplied consumables.
This guide provides instructions for denaturing and diluting libraries after llibrary preparation and before sequencing on the MiSeq. This guide also includes instructions for preparing a PhiX control.
This guide includes information about instrument components, MiSeq Control Software (MCS), and required consumables, and instructions for performing a sequencing run and maintaining the instrument.
MiSeq Reagent Kit Reagent Prep Guides provide an overview of kit contents and instructions for preparing the reagent cartridge for a sequencing run on the MiSeq.